Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I have a question regarding the small board that serves as voltage selector for the mains voltage.
I had a problem on a 248 model - 220V mains when 120V key selected - and opened it up suspecting a blown trafo.
Analyzing the possible voltage selections I noticed that for 220V & 240V mode, one of the windings of the transformer gets shorted (green & brown).
Is this the intended way of going or do I have something messed up in here?
Who is online
Users browsing this forum: No registered users and 1 guest