Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
My 2015 meter has a different offset when using the front jacks 5-6mV and varying while the rear panel is +/- 200nV and relatively stable.
I am moving the same set of test leads from front to rear, so the difference is internal to the meter.
Are there any suggestions on how to easily resolve the issue or is a repair needed?
- Keithley Applications
- Posts: 2845
- Joined: June 10th, 2010, 6:22 am
- Country: United States
Mfg cal adjusts the rear zero..
If front zero is only being done then this could be why the difference.
Who is online
Users browsing this forum: No registered users and 2 guests