Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
We are characterisating OLED and OLEFET devices.
i want to use konika minolta cs100 a with keithley 4200-scs.
The choromometer has hrs connection cable.
What i need is to collect luminance values, source, drain, gate voltage and current values in one excel file together.
i wonder if it possible
Thanks for any ideas
- Keithley Applications
- Posts: 1458
- Joined: October 15th, 2010, 10:35 am
- Country: United States
I took a quick web site look at the Konica product that you mentioned. It seems that it has an RS-232 interface for external command control.
You would have to make use of the 4200-SCS/F extensibility by writing a UTM to communicate with the Konica. There are ksprev and kspsnd commands available to the UTM for RS-232 communications.
You may also consider incorporating commands to control the SMU from this same UTM.
Who is online
Users browsing this forum: No registered users and 4 guests