Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
- Keithley Applications
- Posts: 2845
- Joined: June 10th, 2010, 6:22 am
- Country: United States
If you want to use a material to test the 8009, try using a blank sheet of printer copy paper. Cut it down to fit with-in the 8009 electrode spacing limits ( 21/2 to 4 inches). Test the resistance of paper as baseline for your set-up. Copy paper should have insulating range of 10(8) to 10(11) Ohm-cm for Volume. A table of various insulating materials and their specifications is located in the Keithley Low Level Handbook.
Refer to Low Level Handbook 6th edition page 2-12 Figure 2-10
Who is online
Users browsing this forum: No registered users and 1 guest