Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I am trying to measure current and resistivity (with a voltage bias 1-10V) with the Keithley 2636A.
However, I get negative current and resistance readings.
Why is that so and how to rectify?
- Keithley Applications
- Posts: 1548
- Joined: October 15th, 2010, 10:35 am
- Country: United States
Who is online
Users browsing this forum: No registered users and 1 guest