Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I have two questions related to the interlock violation error during the resistance and resistivity measurement.
for the resistivity measurement, we used the 8009 test fixture. Nowadays we can't measure the sample because the interlock error frequently appears during the measurement. After trying to open and close the lid, and to push and release the top electrode in the fixture several times, we could measure the sample.
Could you let me know what happened to us and how to solve this problem?
for the resistance measurement, we followed the direction as described in 4-20 in your reference manual. we were about to measure the common resistor (10 Mohm, axial type). However, we saw the interlock violation error when we press the OPER button. In order to disappear the message, we tried to connect the interlock cable from 6517b to air (floating) or to 8009. I think our setup is the same as the old post (viewtopic.php?t=99960). However, he/she could measure the sample, but I don't know how he/she connect the interlock.
Could you let me know how to connect the interlock for resistance measurement?
- Keithley Applications
- Posts: 1548
- Joined: October 15th, 2010, 10:35 am
- Country: United States
Similar to common microwave oven, only when door is closed, or 8009 clamshell closed and latched, will the micro switch be able to send the “it is safe to operate” signal on interlock cable.
Have you been using the latch when closing the 8009?
Who is online
Users browsing this forum: No registered users and 1 guest