Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
- Keithley Applications
- Posts: 1436
- Joined: October 15th, 2010, 10:35 am
- Country: United States
Forgetting about the 2602A and in general, to offset it, a Wheatstone bridge type approach would be needed. However, again the stated resolution goals puts extreme demands on the ppm stability of the other resistive elements in the bridge as well as the required voltage resolution of the null detector.
Thermal stability of entire apparatus would be paramount as well.
Who is online
Users browsing this forum: No registered users and 4 guests