Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I am using keithley 2700 multimeter, I have written simple script in C# on .net to get the Resistance measurement readings on RS-232 port.
I have configured the rs-232 port and i have enabled rs-232 port.
I am using the front panel to set measurement settings(i.e. omega2 for resistance) but I need the measurement readings on RS-232 port.
for this I have tried below code:
Just .. READ?
For all of the above codes, I do recieve the response from multimeter but it is un readable.
for eg: I receive: +&)?6??1S?0?5,%S?2i?2??M,?&?4?E9G??
Pls help me with code to get proper readings.
you should check first the data format you have specified. If not specified, please send the following SCPI command FORMat:ASCii to specify it.
Format for transferring data could be ASCII, SREAL, REAL32, REAL64...
To get readable data, you should select ASCii format.
Hope this can help you. Let me know if I can do more.
Who is online
Users browsing this forum: No registered users and 3 guests