Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
Anyone have an example?
Also, can I run a TSP script on my 2450 and have it trigger or communicate with a script running on my DMM7510?
Sorry for the questions, I'm new to the TSP scripting.
- Keithley Applications
- Posts: 1280
- Joined: October 15th, 2010, 10:35 am
- Country: United States
Typically one of the products would use node 1 and would be the master node.
The computer connection would be to this master node (over USB, GPIB or Ethernet).
From this connection, you can access subordinate nodes with the node[X].command type nomenclature.
In addition to basic communication, TSP-Link also provides three lines for triggering. These triggering lines can be used for handshaking or other coordination tasks.
Current product list that supports TSP-Link : 2600 Series of SMUs, 3706A DMM/Switching, 707B Switching, 2450, 2460, DMM7510
Check out the 2450 User Manual, chapter 7, for a detailed treatment of IV characteristics of FETs using two 2450. Both SCPI and TSP commands are provided.
Likewise, the DMM7510 User Manual, chapter 9, details integration of the DMM750 with the 3706A-S switch using TSP-Link.
I've been making good progress on my scripts... now what I need is budget to get a second SMU, perhaps a 2460 this time.
Who is online
Users browsing this forum: No registered users and 1 guest