Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
Download 1.41c (released January 10, 2011)
Download 1.40k (released June 8, 2010)
Download 1.32a (released March 8, 2010)
Download 1.31a (released June 23, 2009)
For the most complete firmware installation instructions, please refer to the release notes included with the firmware update. These may include additional details such as upgrading the instrument's OS.
To install Series 3700 firmware simply place the downloaded .CAB file in the root folder of a USB thumb drive then insert the drive into the front panel of the 3706. If the downloaded file was a .ZIP file, extract from the Zip file the .CAB file before copying it to the USB thumb drive. From the front panel of the 3706 press the MENU button then scroll right until the option UPGRADE is highlighted. Press ENTER. From the Upgrade Unit? menu select YES or PREVIOUS then press ENTER. The upgrade process will commence. Do not use or power off the unit until the upgrade process is complete and the unit has restarted.
Who is online
Users browsing this forum: No registered users and 1 guest