Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
how can I calibrate and verify the frequency and pulse length of the pulse from a function generator? How can I know that the pulse width is not varying slowly, say, over the course of 15 minutes, by, say, 5%? How can I know that the frequency of the pulse train is also not slowly varying by about 5%? How does one make such a measurement?
- Keithley Applications
- Posts: 2849
- Joined: June 10th, 2010, 6:22 am
- Country: United States
Who is online
Users browsing this forum: No registered users and 0 guests