Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
- Posts: 1
- Joined: October 14th, 2011, 5:36 am
- Country: Spain
- Location: Santiago of Compostela, Spain
I've installed the labview 2011 in a new computer with a new Keithley 3A488-LPCI gpib card and now I've a problem with the visa communication with the same instruments. I've tested some "exe" files compiled with the 2009 version and they work fine at this same computer.
If I open in labview 2010 the NI VISA interactive control application I see something like this:
--- GPIB0 (Unknown GPIB board)
--- PXI0 ...
If I make a double click on the gpib 7 instrument I can see the vi_write and read options and I can send commands to the instrument (*IDN?), in this case the keithley 2400 source meter and it answers with the correct information (Keithley....) So all the connections are fine, is it possible that the error cames from this unknown GPIB board message ?.
When I run any project compiled with the 2009 version, they works fine too. But it does not work with the 2011 version, I click on the refresh option of the VISA combos to see the VISA detected devices, no one appears inside, like in the 2009 combos, so the refresh function does not work. The mex explorer does not show the gpib card, is that the problem ?
So, what should be the next step to check the problem ?
Thanks in advance ...
What OS are you using? (windows 7 or 8? 32-bit or 64-bit?)
Is this LabVIEW 2011 32-bit or 64-bit?
I would recommend that you uninstall the KUSB-488B (KI-488) driver and then install the KUSB-488A driver. Here is the link to the driver:
After installing the driver you need to run TRTEST to see if the driver can communicate with your instrument.
Then, if that works run NI VISA Interactive control and attempt to communicate with the instrument.
If all that works, then try LabVIEW again.
(in the future please enter a new forum topic, as you have joined another Customer's topic here and although your problems appear similar, the solution to each may not be the same, and will confuse all involved)
Who is online
Users browsing this forum: No registered users and 1 guest