Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
When I use TekVisa ActiveX control:
Anything above 600ns it fails, and anything below 600ns it works. However, I can manually change it to 800ns.
I don’t understand why this would be the issue?
What driver did you install for you control AFG? How do you open communication between PC and the instrument before sending commands?
Thanks, appreciate any feedback..
Who is online
Users browsing this forum: No registered users and 3 guests