Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
Pardon me if this a very basic question, but I am very new to this field. I just started looking at a procedure which is giving some issues with the C series however it runs fine for B series.
So before I start looking at the procedure I wanted to know about the difference between the AFG 3000 B series and AFG C series.
I looked at the specifications and looks like the procedure has the correct specifications for all the models.
I hope someone can help me!
Thank you in advance!
could you elaborate on the difficulty that you are having with the C series?
But we have tried with 3 different C series assets and it does not work.
It times out whenever we try running the test the first time.
We have to repeat the test again and the second time it will not give any time out error and just runs the test fine.
I am not sure why it is happening. If it was the commands then it should just not do it the second time too.
IEEE-488: (NI EABO) Timeout. I/O operation aborted.
It is definitely got to do with the way it is communicating with the C series.
Hence I just wanted to know what is the difference between the B series and the C series
So we solved the issue using USB instead of the GPIB. It works fine if we use the USB.
Who is online
Users browsing this forum: No registered users and 4 guests