I was wondering, is there any way to programmatically obtain the ip address of an AFG3011C signal generator ?
I've been looking throughout the programmer manueal but i didn't find anything.
Can someone help me please ?
Basically, i'm communicating with the instrument using VISA.
Why i want to obtain it's IP address. Because for DPO/MDO oscilloscopes i will use the ip address in order to save the oscillosope's waveform as png.
I want to do the same thing for AFG3000 series signal generator.
Is it possible ?
- Capture1.PNG (7.05 KiB) Viewed 13639 times
- Capture.PNG (459.42 KiB) Viewed 13639 times
The AFG3000C series only provides provisions for saving screen images to a USB thumb drive plugged into the front panel. This can be done via the front panel controls or by sending the command HCOPy:SDUMp[:IMMediate] to the instrument via the remote interface.
The AFG3000C does not have a web server so there is no web page where you could go to get the screen image.
The only option for saving the screen image of the AFG is to a USB drive plugged into the front panel.
Who is online
Users browsing this forum: No registered users and 0 guests