Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I am using first time two 2614B Keithely with 4 source-measure-units for solution gate FET characterization with reference electrode transfer and output. I am not sure that I did right connection of the four channels to the common ground. I am using the Force LO terminal of all SMU's, connect them together and also to the green banana connectors at the rear of the two instruments. However in the manual I found you should connect the sense LO together. I'm not sure what is correct.
Second if I connect the Force HI to the ground only current is not flowing. Can you please tell me where is the problem?
- Keithley Applications
- Posts: 1436
- Joined: October 15th, 2010, 10:35 am
- Country: United States
For your second question, what are the SMU settings? Forcing what? What compliance limit? Try using some resistance between HI and LO rather than shorting it.
Who is online
Users browsing this forum: No registered users and 1 guest