Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
- Posts: 3
- Joined: August 23rd, 2016, 11:18 am
- Country: United States
I am a bit confused about the measurement range of TPP0850. I noticed that on Tek manual it says it could be use to measure 1000Vrms, but on Tek website https://www.tek.com/datasheet/passive-h ... e-probes-1
says it could be used to measure switching transistor operating at 1200Vrms. So I am confused about the exact voltage range it could measure. Is it 1000Vrms or 1200Vrms?
I want to use TPP0850 together with BNC tip adapter (P/N: 013-0291-xx) to measure the voltage of a device operating at 1kV DC voltage. Is it okay to use them? Thanks!
- Tektronix Applications
- Posts: 544
- Joined: July 31st, 2008, 8:26 am
The maximum voltage for the TPP0850 is 1000Vrms CAT II. The web site has a misprint in the text and will be corrected. As for measuring 1000Vdc, that is not problem for the probe. However, the adapter (013029102) that you are considering using is only rated to 30Vrms. I would use either the large or small hook tips that are supplied with the TPP0850 to connect to your DC source.
Who is online
Users browsing this forum: No registered users and 0 guests