Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
However, READ?\n command does not work at all.
That command only works after I reset the keithley 6485 by *RST\n command.
If I change some settings (like measurement unit or zero correction on-off), then READ?\n command does not work.
Why I can use READ?\n command only after *RST\n command?
I must read data in different setting.
- Keithley Applications
- Posts: 1514
- Joined: October 15th, 2010, 10:35 am
- Country: United States
Is there perhaps some sort of error message? You can certainly change settings and still have it respond to new read commands.
Can you capture a trace file from NI IOTrace? This might let us better understand.
BTW: is this GPIB or RS-232?
Who is online
Users browsing this forum: No registered users and 0 guests