Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I'm trying to measure current with a Keithley 2450 (LabVIEW 8). The current in sleep mode is around 3uA and has a peak at 9mA and with 0.1ms width, I would like to capture the pulse and especially the peak.
I saw others ask about the Max sample rate for Keitely 2450, but I haven't found the answer. I am sending SCPI command in my LabVIEW code, I have set the NPLC to 0.01,
turn off the Auto range and Average, but still I have the interval between samples around 3ms which is too low. Is there any way to increase it?
(And I am using Keithley drivers)
I appreciate any help.
Who is online
Users browsing this forum: No registered users and 0 guests