Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
Questions in this forum area are community supported. Tektronix does not regularly monitor posts in this area.
- Posts: 1
- Joined: January 27th, 2021, 10:28 am
- Country: United States
I ended up getting a “tested” 2445 scope and I’m having an issue I’ve never seen before, it runs fine for a few min and does this once or twice and then be fine for a while longer. Also the calibrator output doesn’t look to be working, anyone run into this before? Thanks
https://previews.dropbox.com/p/thumb/AB ... ize_mode=5
Who is online
Users browsing this forum: No registered users and 4 guests