Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I have tried to connect a sourcemeter 2410 to my computer by a GPIB but it doesn't work. I have used keithley flash wizard32. This program has detected what is connected to the gpib but when i have to choose the file ()2400c32, there is an error "model 2410 not connected".
This gpib card works with others Keithley like : 485 autoranging - ; 705 scanner ; etc...
Can you help me please ?
Check that you have chosen GPIB communication, not RS-232, in the 2410. If you are using a NI GPIB card, you can use Measurement & Automation explorer to scan for the instrument and correspondingly communicate with it. Same applies for Keithley GPIB-488A. You can send a query *IDN? and read back the response.
If this doesn't work, u can switch the communication to RS-232 and try communicating it using hyper terminal. If it works using RS-232, then the GPIB card in the instrument might be faulty. In this case, do contact the service center in your region to get the instrument repaired. Thanks.
Who is online
Users browsing this forum: No registered users and 3 guests