I need to set an 2015P DMM to 10 Hz Output frequency.
If i do this manually, i get an Output of 10 Hz.
If i want to set 10 Hz Output frequency via Labview i get an Error Message.
In the documentation of the Ke2015.dll i read that it is only possible to set an output frequency of 10 Hz on 2016 Models?
Is there any Chance to set an 2015 to 10 Hz also?
Why is it possible to set 10 Hz manually and why is it not possible to set it via dll call?
Is there an new dll available? The driver seems to be version 22.214.171.124 (Dated on Feb 1st, 2010)
- Keithley Applications
- Posts: 2847
- Joined: June 10th, 2010, 6:22 am
- Country: United States
And it accepted this command.
Then I sent
And I could read the 10Hz wave on an oscope.
I was not able to open the driver as it is too old.
Try sending just these commands and see if that helps.
Who is online
Users browsing this forum: No registered users and 0 guests