Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I tried Kickstart, but it seems to be limited to a single DMM7510.
Is there a way to synchronize two DMM7510 and get continuous VDC samples from both units?
There's no voltage trigger in this setup, but I would settle for stopping VDC measurements on both units simultaneously and writing down the VDC measurement value. If that's the route, how can I stop both units at the same time?
Attached are two .tsp files which set up each instrument for synchronizing measurements. Note this is a very basic application which has some variation between reading times however for the most part each instrument is acquiring a reading ~6 microseconds apart from one another.
Who is online
Users browsing this forum: No registered users and 0 guests