Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
Since I'm displaying the readings on a temperature vs. time graph, the result is a weird spike near the beginning of every run. I've considered simply throwing away the first couple of scans, but I'd much rather address the root of the problem if possible. Any suggestions would be appreciated. the TSP commands I'm using are listed below. Thanks! -AW
Commands sent once at the beginning to set up parameters.
dmm.func = dmm.TEMPERATURE
dmm.autodelay = dmm.OFF
Commands sent every 500ms
Try turning it back on and see if your measurements look OK after.
Who is online
Users browsing this forum: No registered users and 2 guests