Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I have to perform a continuous measurement with a Keithley 2750 DC Voltage with 40 channel / second measurement. For this I would like to use the internal memory of the 2750. What I have not done before. Based on the available examples, I have copied different parts to a new program. However, it does do the work as expected. Not all channels were measured and the transferred memory values I am not sure that they are correct. Maybe someone knows himself with the buffer of the Keithley 2750/2701 and can help me.
- Keithley 27XX Performance Buffer.vi
- (36.51 KiB) Downloaded 654 times
As I received no answers to my post, I was ongoing with labview.
In the current Vi I do not adress any buffer.
The keithley 2750 does use the internal automaticly.
I have this vi in a while loop and it works.
What is the advantage of addressed buffer?
- Keithley 27XX SMBT.vi
- (38.26 KiB) Downloaded 476 times
Who is online
Users browsing this forum: No registered users and 0 guests