Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
Can any one kindly tell me if I can use KI2701 to measure IV curve from a solar simulator?
Please tell me how if it is possible.
If not, what about KI487? Specifically kindly tell me what do I require to do so.
- Keithley Applications
- Posts: 1547
- Joined: October 15th, 2010, 10:35 am
- Country: United States
The 487 is a - with a DC voltage source.
The 2701 is definitely not suitable to perform IV curve.
If currents from your solar device are within the capability of the 487 - and you make a custom cable to combine the DC source with the AMPs measure function, then it can maybe play a role. But your currents will need to be pretty small.
A SourceMeter is the best product for IV curves. Check out the model 2450.
Who is online
Users browsing this forum: No registered users and 0 guests