Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
It only allows to carry out measurements over 37 frequency from 1kHz to 10 MHz. I understand that calibration is probably done only at
certain locations, but that can always be interpolated to increase number of frequency samples.
What can I do to increase the number of data points in this C-f sweep.
• 1 kHz to 10 kHz in 10 kHz steps
• 10 kHz to 100 kHz in 10 kHz steps
• 100 kHz to 1 MHz in 100 kHz steps
• 1 MHz to 10 MHz in 1 MHz steps
Who is online
Users browsing this forum: No registered users and 0 guests